Sonication optimization using Cup horn - (Jun/10/2009 )
I am conducting a Chip assay with HepG2 cell, following the Upstate Chip assay kit protocol.
It's my first time to do this experiment, so I guess I might have to go through many trials and errors.
But I wish I could find anyone who can help me minimizing that.
My problem occured at sonication step.
I am using Vibra Cell sonicator VCX-500/750 (Sonics) with a cup horn.
Before sonication, I aliquoted the resuspended cell with 1X10^6 cell / 200ul concentration in eppendorf tube.
Sonication was performed at 22kJ ~ 100kJ energy with 5 X 10 s, 10s interval, or 5X 30s, 30s interval. All at 30% amplitude.
However, the sheared DNA sizes of different samples were almost same; the average size was approximately 1200 bp.
My question is :
1. What is your advice? Should I raise the sonication energy? or time? which would be more conclusive?
2. If you are using a cup horn method, can you provide me your own sonication condition?
(especially with your sonication energy and time)
i don't use cup horn, however i've found it helps to try sonicating 1X10^6 cell / 50 microliters concentration in eppendorf tube.