way to know if my UNOII pcr machine is working properlly
Posted 17 July 2009 - 11:11 PM
i am working now for two months doing pcr and doesnt manage to get reliable results, It seems that if i am using pcr of another lab i get a band but if i am using my ocr machine i don't. is there a way to self check my UNOII biometra pcr machine?
Posted 18 July 2009 - 06:15 AM
Seems like you may have already done what I would suggest first -- make a complete PCR reaction mix (primers, template, dNTPs, MgCL2, etc.) that is twice the volume you'd normally use. Split this mixture equally amongst two tubes. Run one tube in your machine, and the other the machine from the other lab, using the exact same PCR program in both machines.
If you get an amplicon from the other lab's machine, but not from yours, there's likely something wrong with your machine.
In your machine, set a program for the block to cycle between something like 4C and 37C, 10 seconds each loop. Run the program and feel the block with your hand -- is the temperature changing?
Do you have a manual for your machine?
Posted 20 July 2009 - 08:16 AM
My two cents
Posted 20 July 2009 - 01:44 PM
A good point -- if your PCR is marginal, perhaps selecting new primers would help?