Well, I left my pH probe in distilled water over the weekend and now it doesn't work anymore. The pH is registering at between 2 and 3 whatever standard is measured, ie 4,7,10. I put the probe in saturated KCl overnight but very little improvement. Have I destroyed it or is there anyway I can get it back on track? Thanks for the help, FreddieP
Destroyed my pH probe
Posted 27 October 2013 - 10:02 AM
Depending on the probe, there may be a way to refill the interior solution of the probe with saturated KCl, which would likely restore its function.
Posted 27 October 2013 - 03:23 PM
Depends on the probe as phage434 says. The probe I use in the lab has to be stored with a sponge wetted with MQ, never KCl... anyway, try to change the internal solution, rinse it with some electrode solution before start to testing. Calibrate as usual and test.
Posted 28 October 2013 - 11:37 AM